In this paper, a femtosecond two-photon-absorption laser-induced-fluorescence (fs-TALIF) diagnostic was designed, installed, and operated on the Princeton-Field-Reversed Configuration-2 device to provide non-invasive measurements of the time and spatially resolved neutral-atom densities in its plasmas. We demonstrated that fs-TALIF can provide spatially, to ±2 mm, and temporally resolved, to 10 µs, measurement of the density of certain previously inaccessible atoms, e.g., atomic hydrogen (Ho).
Calibration of the Ho density was accomplished by comparison with Krypton (Kr) TALIF. Measurements on plasmas formed of either molecular hydrogen (H2) or Kr fill gases allowed examination of nominally long and short ionization mean-free-path regimes. With multi-kW plasma heating and H2 fill gas, a spatially uniform Ho density of order 1017 m−3 was measured with better than ±2 mm and 10 µs resolution. Under similar plasma conditions but with Kr fill gas, a 3-fold decrease in the in-plasma Kr density was observed.
Ho density is essential to several plasma diagnostics including time-of-flight and ion energy analyzers, and high-resolution spectroscopy, as by CPT (coherent population trapping) and DFSS (Doppler-free saturation spectroscopy). It was also used in Collisional Radiative Modelling for predicting the Electron temperature diagnostic in PFRC-2.
This paper talks about a collisional-radiative (CR) model that extracts the electron temperature, Te, of hydrogen plasmas from Balmer-line-ratio measurements and is examined for the plasma electron density, ne, and Te ranges of 1010–1015 cm−3 and 5–500 eV, respectively. The first tests of the CR model on the Princeton Field Reversed Configuration-2 (PFRC-2) have been made, including comparisons with other diagnostics. These comparisons are informative as different diagnostics sample different parts of the electron energy distribution function.
The above media was taken when we were running with the Rotating Magnetic Field (RMF) Heating System. The video from 10 to 23 seconds shows the plasma rotations are more pronounced and then stabilized later on. The stabilization of plasma is due to the gas puff introduced.
While the PFRC is under upgrade to lower the RF frequency, we have been running the seed plasma, which is PFRC plasma operation without RMF. We also observe rotation in the PFRC seed plasma.
The Princeton Field Reversed Configuration-2 (PFRC-2) upgrade in field and frequency is underway. We are currently installing new coils around the experiment to increase the magnetic fields and new capacitors to help lower the RF operating frequency – all to reach our target milestones of measuring ion heating! This is an essential next step in our development of Direct Fusion Drive.
The power supplies are stacked in their rack, ready to supply power to the belt coils. The supplies must be programmed to energize for each pulse as they are not cooled and the coils would otherwise overheat. The belt coil holder component on the right was 3D printed at PPPL.
The new 2 nF capacitors, shown above (left image), must be enclosed in a custom copper box that will be part of the tank circuit of PFRC-2. Each component must be carefully designed, including the lengths of the connecting cables, for us to get the right frequency without exceeding voltage limits of the materials.
The above image is of the cable that will connect the tank circuit and the PFRC-2. These cables are very robust, and stiff so that the layout must be carefully planned. We will continue to post updates as we work towards that 2 MHz frequency milestone!